Lab supply | First online laboratory shop, and selling where new features were introduced. practice. Instrumental factor must be checked. Chema Diagnostica will not be responsible for instrumental bad programming. Mindray BAA. Mindray) owns the intellectual property rights to this Mindray product and this manual. This manual may refer to information protected by copyrights or patents.
|Published (Last):||21 September 2004|
|PDF File Size:||4.53 Mb|
|ePub File Size:||1.26 Mb|
|Price:||Free* [*Free Regsitration Required]|
Chemistry Analyzer BA-88A Mindray medical equipment
Medicines, anticoagulants or preservative in the samples may lead to unreliable results. System Specifications System specifications Dimension: Up to tests can be programmed 6. Troubleshooting Troubleshooting This chapter presents all warning messages and recommended corrective actions, which should be taken in time once any error occurs.
Comments to this Manuals Your Name. Detection Of The Input Signal 4. MCU and FPGA outputs motor control signal, which can control the step motor of the filter wheel via step motor driver circuit. This manual may refer to information protected by copyrights or patents and does not convey any license under the patent rights of Mindray, nor the rights of others.
Page 65 Locate the position where the peristaltic pump is installed. When disassembling, if it is not necessary to replace the motor, filter wheel or sensor stop plate, we recommend you to keep the connection between the minsray and the filter wheel, thus the installing procedure would be simplified; Sectra enters South Korean market by signing digital pathology distribution agreement with HuminTec.
MINDRAY BAA SERVICE MANUAL Pdf Download.
Page 11 Treating Waste Liquids Please observe the following instructions to prevent environmental pollution and personal injury caused by waste. Replacing Cpu Board Pull the wire out mundray you can see the wire joint connecting the lamp assembly, with one end connecting to the lamp and the other end connecting to the inside of the instrument.
Install pump head mindrah the pump, with motor shaft into installation hole Pinch buckles to lock the pump head After installing the pump head, connect the tubing with the tubing that goes through the backboard and the waste tubing. If necessary, you can disconnect the connection of some cables in midnray.
The logs will record mindrayy time, level, mindeay and detailed information of each warning to help user record and search errors. Page 5 Company Contact Manufacturer: Relevancy Transaction Level Response Rate.
Mindray – BAA Community, Manuals and Specifications | MedWrench
This manual is geared for service personnel authorized by Mindray. Figure System structure of the software 5. Get best deals for coconut.
The configuration procedures are shown as follows: Mindrzy the instrument is in use, if you notice that the instrument can not aspirate or aspiration volume decreases mkndray, check for leakage in the peristaltic pump.
The last day of the fair at this year’s Medica was also the last Pull out the tubing that goes through the backboard of the analyzer for mm until the adapter is exposed. The Ethica Award, the highest honour of the European cardiovascular Page 10 Preventing Personal Injury Caused by Photometer Lamp Please observe the following instructions to prevent personal injury caused by photometer lamp.
Please service the system strictly as instructed by this manual. Driver Of The Step Motor 4.
Click Maintenance on the main screen of the instrument. Product Specification Automation Grade Manual.
Prepare the samples, reagents and distilled water;